Ultra high pressure mixers are sold as complete assemblies only.
Delay volume specified below was calculated based on an ASI engineering drawing. The delay volume can be verified by injecting a sample at a known flow rate, then measuring the time this sample first appears at the exit of the mixer.
Back pressure varies directly with flow rate. For example, if the back pressure is 2 PSI at 1 mL/min., it will be .2 PSI at 100 µL/min. At flow rate less than 100 µL/min., the back pressure is negligible.
Port size: 10-32 CPI (standard HPLC fittings for 1/16″ tube)
Compatible with Thermo Scientific™ nanoViper™ Fingertight Fittings
Maximum Operating Pressure: 18,000 PSI
Delay volume:10 µL
Back pressure: < 2 PSI @ 1mL/min. with water