High Flow Series cartridges:800 µL, 1 mL and 1.5 mL are interchangeable and fit in all High Flow Series In-line, Binary tee and Ternary tee housings..
Delay volume specified below was calculated based on an ASI engineering drawing. The delay volume can be verified by injecting a sample at a known flow rate, then measuring the time this sample first appears at the exit of the mixer.
Back pressure varies directly with flow rate. For example, if the back pressure is 2 PSI at 1 mL/min., it will be 4 PSI at 2 mL/min.
Currently we offer only In-line biocompatible housing for 800 µL, 1 mL and 1.5 mL series. If you need Binary mixing, you may add 0.040″ or 0.050″ thru hole PEEK Tee for 1/16″ OD tubing (for example, IDEX Part number: P-728).
Port size: 10-32 CPI (standard HPLC fittings for 1/16″ tube)
Compatible with Thermo Scientific™ nanoViper™ Fingertight Fittings
Maximum Operating Pressure: 3,000 PSI
Delay volume: 1.0 mL
Back pressure: < 2 PSI @ 1 mL/min. with water